Thermo Scientific Ezy-Cal Count Precision Size Standards
Meet the unique and demanding requirements of the contamination control laboratory in modern industries.
Manufacturer: Thermo Scientific 6015
Meet the unique and demanding requirements of the contamination control laboratory in modern industries with Thermo Scientific Ezy-Cal Count Precision Size Standards, easy-to-use liquid particle count controls at 2000 particles per mL. Designed for regular use in liquid particle counters to verify instrument calibration before each use, standards assure reproducibility of the particle counter by permitting a continuous record of its performance using a NIST-traceable microsphere suspension with a precise concentration.Features:
- Ready-to-use (no dilution required) particle size standard for validating optical particle counters
- Magnetic stir bar included in each bottle for clean, convenient and direct sampling
- Aqueous suspension medium contains a combination of dispersing agents to keep particles from clumping or sticking to flow surfaces in the particle counter
- Particle diameters traceable to the Standard Meter through the National Institute of Standards and Technology
Checking the calibration of liquid particle counters.
|Contains trace amount of surfactant|
|2000 particles/mL ± 10%|
|Used for validating the performance of liquid particle counters|
|15.02 ± 0.15µm|
|s = 0.15µm, 1.0% CV|
|Unless otherwise stated, refrigerate 2° to 8deg;C product when not in use but do not freeze. Store upright and keep bottle tightly sealed. Mix product with gentle inversion by hand or vortex mixer.|
|Liquid Particle Counter Calibration|
|Polystyrene microspheres in water|
|Certificate of Calibration and Traceability to NIST|
|1.59 at 589nm (25°C)|
|Product stable for a minimum of three months|
|Count Control Particles|
We continue to work to improve your shopping experience and your feedback regarding this content is very important to us. Please use the form below to provide feedback related to the content on this product.
Your feedback has been submitted. Fisher Scientific is always working to improve our content for you. We appreciate your feedback.Ok