Thermo Scientific™ National Scientific UV Detector Performance Validation Kit
Manufacturer: Thermo Scientific™ CK09001S
Tests detector precision, accuracy, and linearity in less than 30 minutes
Verifies performance of UV-visible detectors in UV spectrophotometers, HPLC, and CE detectors. Confirms detector wavelength settings and linear response when validating new methods—ideal for methods using extinction coefficients or factors. Quick, three-step method measures absorbance linearity, wavelength accuracy, and wavelength precision. Absorbance linearity. Kit uses holmium oxide solution as wavelength reference standard. Calibrates at 241, 278, 287, 361, 451, and 537nm. To check wavelength accuracy, fill cell with holmium oxide solution and measure absorbance ±5nm from target wavelength. Wavelength error is the difference between observed wavelength at maximum absorbance (λmax) and target wavelength.Wavelength precision. To determine wavelength precision, calculate wavelength accuracy, then set detector wavelength at least 10nm from target wavelength. Reset wavelength to target wavelength and record absorbance. Wavelength precision equals the relative standard deviation of five measurements.
Replacement solutions are available; contact your Customer Service Representative.
|UV Detector Performance Validation Kit|
|Verifies Performance of UV-Visible Detectors in UV Spectrophotometers, HPLC and CE Detectors|
|Eight 30 mL absorbance solutions, one 30 mL wavelength calibration solution, one 60 mL absorbance reference solution, Microsoft excel and lotus template disk, syringes and HPLC detector connection kit and certificate of analysis|
We continue to work to improve your shopping experience and your feedback regarding this content is very important to us. Please use the form below to provide feedback related to the content on this product.
Your feedback has been submitted. Fisher Scientific is always working to improve our content for you. We appreciate your feedback.Ok