Thermo Scientific Orion 900100 Sure-Flow Reference Half-Cell Electrode
Precisely measure with ion-selective electrodes using this sleeve-type Ag/AgCl reference electrode.
Manufacturer: Thermo Scientific 900100
Precisely measure with ion-selective electrodes using the Thermo Scientific Orion 900100 Sure-Flow Reference Half-Cell Electrode, a sleeve-type Ag/AgCl reference electrode.
Reference electrode Orion 900100 is recommended for all pH and ion-selective measurements except for those:
- In dilute solutions where the ion to be measured is contained in the filling solution of the Orion 900100
- In solutions where an ion in the filling solution would introduce an electrode interference
- In solutions where a species is present which forms a precipitate with an ion contained in the filling solution of the Orion 900100
For these samples, the double junction reference electrode, Orion 900200, with an appropriate intermediate filling solution, is recommended.
When filled with Orion 900001 Filling Solution, the Orion 900100 matches the potential characteristics of a conventional KCl saturated calomel electrode.
- Many ISE determinations
- Routine pH measurements with Orion 910100 Electrodes
|0° to 100°C|
|Equitransferent solution with Ag/AgCI|
|900100 Sure-Flow Reference Half Cell Electrode|
|Orion 900001 Filling Solution|
|Sure-FlowSureFlow sleeve junction|
|0 to 14pH|
|Single Junction Ref Electrode|
We continue to work to improve your shopping experience and your feedback regarding this content is very important to us. Please use the form below to provide feedback related to the content on this product.
Your feedback has been submitted. Fisher Scientific is always working to improve our content for you. We appreciate your feedback.Ok
Limited warranty period: Subject to all warranty requirements and exclusions set out in our Terms and Conditions of Sale, this product is warranted from the date we ship the product and for ninety (90) days thereafter. Disclaimer to internal research use restriction: Any restrictions on the purchaser to utilize the product(s) for internal research purposes only does not apply to the product(s) on the web pages associated with this statement.