Thermo Scientific DXT-RAD Extremity Dosimeters and Accessories
Disk dosimeters with novel and permanent individual barcodes feature hot or cold sterilization with fast readout
Manufacturer: Thermo Scientific 28834
DXT-RAD is suitable for extremity dosimetry applications.
- Sealed disposable finger ringssterilized hot or cold
- Ring sealing and chip extraction
- Carrier cards for readout after exposure
All items are for use with DXT- RAD extremity dosimeter, TLD materials.
Item 26994 is compatible with Modelsl DXT- 100, TLD-100; 100mg/cm2
Item 26995 is compatible with Models DXT- 107H, TLD- 100H; 7mg/cm2
Item 26996 is compatible with Models DXT- 600, TLD- 600; 100mg/cm2
Item 26997 is compatible with Models DXT- 600, TLD- 600; 100mg/cm2
Item 26998 is compatible with Models DXT- 707H, TLD- 700H; 7mg/cm2
Item 26999 is compatible with Models DXT- 760H, TLD- 700H; 63mg/cm2
Item 500597: 42mg/cm2
Item 27176: 5mg/cm2
Item 28573: 3.3mg/cm2
Item 500596 is yellow.
Item 500608 is orange.
Item 500609 is lavender.
Item 500624 is mint.
Wristbands and holder
500644 wristband holder is orange.
500640 wristband is braided white.
500641 wristband is solid green.
500642 wristband is solid brown.
500643 wristband is solid green.
500598 holds one to four ringlets per card and is gold.
500598-2 holds one to four ringlets per card and is silver.
28834 Ring and cap loader/unloader station includes manual carrier card loader/unloader.
28919 Loader/Unloader is for use with manual carrier card.
26591-1: 115V AC/60Hz
26591-2: 220V AC/50Hz.
Bar Code Readers
These read the assembled ringlet, cap and ring assemblies, and are used in ring issuance and receiving applications.
Electrical requirements: 27196-1: 115V AC/60Hz U.S. plug; 27196-2: 220V AC/50Hz European plug; 27196-3: 220V AC/50Hz U.K. plug.
Handling - Vacuum tweezers should always be used. (Avoid mechanical tweezers or fingers). Small scratches, loss of mass or foreign deposits affect light emission.
Cleaning - Rinse the dosimeters in analytical grade anhydrous methyl alcohol between normal uses. (Do not soak). Dry by leaving to evaporate for at least one hour. Anneal once before actual use, accurately following the established procedure. The anneal will also assist in removing any residual methyl alcohol.
Annealing - For annealing temperatures up to 400°C, the containers should be made from high temperature stainless steel or oxidized aluminum, preferably thin to assist rapid cooling following annealing. (Do not use non-oxidized aluminum). The use of a dedicated annealing oven reduces the risk of contamination by foreign material. Place the annealing containers on open oven racks with air space all round to avoid inconsistent heat gradients. (Do not stack containers or allow them to touch the oven walls).
This product(s) resides on a Fisher Scientific GSA or VA contract. If you are viewing this page as a nonregistered user, the price(s) displayed is List Price. To view your GSA or VA contract pricing, log in using your account number, or become a registered user by contacting one of our Customer Service teams. You can also view your contract price by searching for this item(s) on GSA Advantage. To place an order, contact Fisher Scientific Customer Service.